This article has Open Peer Review reports available.
Misleading measures in Vitamin D analysis: A novel LC-MS/MS assay to account for epimers and isobars
© Shah et al; licensee BioMed Central Ltd. 2011
Received: 29 January 2011
Accepted: 14 May 2011
Published: 14 May 2011
Recently, the accuracies of many commercially available immunoassays for Vitamin D have been questioned. Liquid chromatography tandem mass spectrometry (LC- MS/MS) has been shown to facilitate accurate separation and quantification of the major circulating metabolite 25-hydroxyvitamin-D3 (25OHD3) and 25-hydroxyvitamin-D2 (25OHD2) collectively termed as 25OHD. However, among other interferents, this method may be compromised by overlapping peaks and identical masses of epimers and isobars, resulting in inaccuracies in circulating 25OHD measurements. The aim of this study was to develop a novel LC-MS/MS method that can accurately identify and quantitate 25OHD3 and 25OHD2 through chromatographic separation of 25OHD from its epimers and isobars.
A positive ion electrospray ionisation (ESI) LC-MS/MS method was used in the Multiple Reaction Monitoring (MRM) mode for quantification. It involved i) liquid-liquid extraction, ii) tandem columns (a high resolution ZORBAX C18 coupled to an ULTRON chiral, with guard column and inlet filter), iii) Stanozolol-D3 as internal standard, and iv) identification via ESI and monitoring of three fragmentation transitions. To demonstrate the practical usefulness of our method, blood samples were collected from 5 healthy male Caucasian volunteers; age range 22 to 37 years and 25OHD2, 25OHD3 along with co-eluting epimers and analogues were quantified.
The new method allowed chromatographic separation and quantification of 25OHD2, 25OHD3, along with 25OHD3 epimer 3-epi-25OHD3 and isobars 1-α-hydroxyvitamin-D3 (1αOHD3), and 7-α-hydroxy-4-cholesten-3-one (7αC4). The new assay was capable of detecting 0.25 ng/mL of all analytes in serum.
To our knowledge, this is the first specific, reliable, reproducible and robust LC-MS/MS method developed for the accurate detection of 25OHD (Vitamin D). The method is capable of detecting low levels of 25OHD3 and 25OHD2 together with chromatographic separation from the co-eluting epimers and isobars and circumvents other instrumental/analytical interferences. This analytical method does not require time-consuming derivatisation and complex extraction techniques and could prove very useful in clinical studies.
Vitamin D3 is formed from its precursor 7-dehydrocholesterol in the skin by ultraviolet B light (medium wavelength, 290-315 nm) and Vitamin D2 originates from dietary sources together with some fraction of D3. In the liver, Vitamins D3 and D2 undergo hydroxylation reactions catalyzed by 25-hydroxylase, which leads to the formation of pharmacologically active metabolites 25OHD3 and 25OHD2 respectively (collectively termed as 25OHD). Further metabolism (in the presence of 1α,hydroxylase) in the kidney produces the pharmacologically active metabolites 1-alpha,25-dihydroxyvitamin-D3 (1α,25(OH)2D2) and 1-alpha,25-dihydroxyvitamin-D2 (1α,25(OH)2D3) along with the minor metabolite 24,25(OH)2D3 [8–10].
Since 25OHD has significant effects on health and wellbeing, there has been a substantial interest in improving the relevant analytical techniques [11–30]. Owing to a long serum half-life, measurement of total 25OHD (25OHD2 and 25OHD3) is the routinely used approach for assessing the total circulating Vitamin D status [10–14]. In immunoassay techniques, a measure of total metabolite concentration and equivalent detection of both 25OHD2 and 25OHD3 is challenging, as binding proteins show a higher affinity for 25OHD3 than 25OHD2 [15–18]. Reports have shown inter-laboratory and inter-method variations in results for Vitamin D determinations [19–21].
LC-MS/MS is currently the best technique available for the correct quantification of 25OHD3 and 25OHD2 [22, 23] and it also has the capability to overcome most of the problems associated with protein binding assays. LC-MS/MS is a more favourable technique because sample derivatisation is not required, run time is very short and an internal standard is used which usually compensates for any matrix related and instrumental effects [24–32].
The epimerization of 25OHD3 and 1α,25-(OH)2D3 results in the formation of 3-epi-25OHD3 and 3-epi-1α,25(OH)2D3 epimers respectively as shown in Figure 2. The epimers of 25OHD differ in configuration at third carbon atom (C-3) (shown by dashed highlights in Figure 2) that is attached to a hydroxyl group. Hydroxylation of 3-epi-25OHD3 forms 3-epi-1α,25(OH)2D3 .
The aim of this study was to develop a novel LC-MS/MS method that can accurately identify and quantitate 25OHD3 and 25OHD2 and chromatographically separate epimers and isobars.
Materials and methods
Blood samples (100 mL) were obtained from 5 healthy, male, Caucasian volunteers of age ranging from 22 to 37 years, weight ranging from 72.1-98.1 kg (mean weight 84.86 ± 11.3 kg) and height ranging from 165-190 cm (mean height 179.5 ± 8.6 cm). Blood samples were centrifuged at 1500 g and serum was extracted. The serum samples were frozen individually in labelled, amber, plastic vials . To minimise assay variations, a multilevel serum calibrator set (Chromsystems, Germany) was used for preparation of calibration curves and quality controls. The lyophilised calibrators (based on human serum) consisted of 3 high and 1 low-level calibrators and were handled in the same manner as volunteer specimens. According to the assay procedures, the calibrators were analysed along with routine samples to meet the standards outlined by National Institute of Standards and Technology (NIST) [44, 45].
Standards and reagents
GV-65C (3 mL syringe) mixed bed cation exchange columns were obtained from Biochemical Diagnostics, (New York, USA), Bond Elut-SI, Bond Elut Plexa, Bond Elut LMS, Bond Elut PPL, SampliQ OPT and SampliQ DVB solid phase extraction cartridges were purchased from Agilent Technologies (Cheshire, UK). 25OHD3, 25OHD2, 1-α-hydroxyvitamin-D3 (1αOHD3), 3-epi-25OHD3, hexane, isopropanol, methanol, dichloromethane, deionised water, formic acid, acetonitrile, ammonium hydroxide, pentane and ether were obtained from Sigma Aldrich (Poole, UK). 7-α-hydroxy-4-cholesten-3-one (7αC4) and stanozolol-D3 (internal standard) were obtained from LGC standards (Teddington, UK). All chemicals and reagents were of HPLC grade.
Preparation of standards and samples
Stock solutions of all analytes were prepared in methanol to obtain a concentration of 1 mg/mL and stored in amber vials at -20°C in the dark. Under these conditions, the stock solutions were found to be stable for 3 months. The solutions were kept in the dark to minimise light induced degradation of Vitamin D [46, 47]. Working solutions were made in methanol by serial dilution of stock solutions. Working internal standard solution was also prepared by diluting the stock solution of internal standard with methanol to a final concentration of 1 μg/mL. Calibrators and internal standard solutions made in-house were stable for 2 weeks when stored at -20°C. The lyophilised serum calibrators were reconstituted in HPLC grade water and allowed to stand for 10 to 15 minutes at room temperature. The vials were then swirled to dissolve the contents until homogeneity. The lyophilized reference serum calibrants were stable for 3 months when stored at -20°C [46, 47]. The calibration curves and quality controls were prepared from the multilevel reference calibrator set in the range 0.5 to 84.4 ng/mL concentrations, respectively. Different methods and conditions for sample pretreatment and extraction were undertaken to optimise recovery, specificity and signal to noise ratio. The sample pretreatment and extraction methods [12–32] were adopted as follows.
Serum samples were thawed; vortex mixed and equilibrated at room temperature for 15 minutes and 25 μL of working solution of stanozolol-D3 (internal standard) was added to all samples. Formic acid (2 M, 50 μL) was added and the resultant solution vortexed. Then 3 mL of methanol/isopropanol (1:1, v/v) mixture was added and vortexed to release the protein bound analyte and to promote protein precipitation during a 15 minutes incubation at 4°C. The suspended matter was removed after centrifugation at 3500 g for 5 minutes at 4°C. The supernatants were transferred to clean amber glass tubes. The remaining solution was subjected to further sample purification, as follows.
Liquid-Liquid extraction (LLE)
Different solvent mixtures were tested for liquid-liquid extraction: namely; heptane, methanol, propanol, dichloromethane, acetonitrile and hexane. Hexane/dichloromethane (1:1, v/v) mixture was found to give optimum extraction recovery. For liquid-liquid extraction a 3 mL hexane/dichloromethane mixture (1:1, v/v) was added to the supernatant after protein precipitation and the solution was vigorously vortexed for 1 minute and then centrifuged at 3500 g for 5 minutes at 4°C. The supernatant layer was transferred to clean, amber, glass tubes. The residual lower layer of the serum sample was further extracted twice with the hexane/dichloromethane (1:1, v/v) mixture. The organic phase obtained was pooled and dried under a gentle stream of nitrogen at room temperature. It was then reconstituted in 200 μL of HPLC grade methanol/water (1:1, v/v) mixture. Solid phase extraction (SPE) was also tested as an alternative to the liquid-liquid extraction technique.
Solid phase extraction (SPE)
The mass spectrometer was operated in positive electrospray ionisation (ESI) mode at a spray voltage of 5000 V and capillary temperature of 450°C. The generated protonated molecules of 25OHD3, 25OHD2, 3-epi-25OHD3, 1αOHD3, 7αC4 and stanozolol-D3 (internal standard) were used as precursor ions for collision activated dissociation (CAD) into product ions in MS-MS analysis.
Summary of assay validation results.
Linear range (ng/mL)
Recovery (%) at 50 ng/mL
Under- or over-estimation of actual 25OHD3 concentrations may occur owing to co-eluting epimers (e.g.3-epi-25OHD3) and isobars (e.g.7αC4) [39–42]. The use of high resolution microbore ZORBAX column in tandem with chiral ULTRON column not only facilitated accurate determination of the analyte ions but also chromatographically separated all isobars and epimers from 25OHD co-eluting peaks.
Retention times, MRM transitions and ESI conditions of analytes.
Retention time (min)
Transition (m/z) Precursor→ Product
Collision energy (eV)
401.3 → 383.1
401.3 → 365.1
401.3 → 159.2
413.3 → 395.5
413.3 → 377.2
413.3 → 355.5
401.3 → 383.1
401.3 → 365.1
401.3 → 159.2
401.3 → 383.1
401.3 → 365.1
401.3 → 159.2
401.3 → 383.1
401.3 → 365.1
401.3 → 159.2
332.2 → 81.2
Vitamin D serum results.
The percentage of interfering epimer (E) and isobars (I) in serum.
These results confirm the complexities involved in measuring Vitamin D status and reinforce the need for the uniform adoption of improved accurate assays. Measurements of 25OHD in human serum using competitive immunoassays are difficult owing to lipophilicity/tight binding to the Vitamin D-binding protein (DBP), even at very low serum concentrations . In addition, immunoassays for 25OHD have been reported to cross-react with 24,25(OH)2D3, an intermediate product during the formation of 1,25(OH)2D3 in the kidney [16, 17] (Figure 1). Many commercial immunoassays can only measure 25OHD3 and are not suitable to monitor supplementation with Vitamin D2, which is derived from plant sources and widely used in many countries for fortification of foods . Hence, LC-MS/MS analysis is preferred but it has been shown that it is also subjected to interference issues [19–28]. In our new method, the use of an electrospray ionisation technique has overcome the problems associated with in-source transformation of 25OHD3-sulfate and other metabolites to 25OHD3 which may occur when using atmospheric pressure chemical ionisation mode .
The use of tandem column technology, an optimised mobile phase composition and a modified extraction method not only separate the epimers and isobars but also eliminates the interferences caused by early eluting salts (e.g. sodium) and the late eluting phospholipids, which may interfere with analyte ionisation [29–34]. Moreover, the installation of a low dispersion inlet filter together with a chiral guard column also minimises the interferences resulting from early eluting amino acids and late eluting xenobiotics. The column also separates interfering compounds present at higher concentration which have been shown to overlap or share exact mass transitions with 25OHD e.g. detergents and phthalates etc [29–31].
The separation of epimers and isobars from the target analyte is critical because they can overlap 25OHD peaks and form the same masses upon ionisation, thus compromising the true status of 25OHD in circulation [38–43]. The epimer of 25OHD3 is known to have the same effects on suppressing parathyroid hormone (PTH) secretion, but it has negligible calcium-producing effects [40, 41]. Recently, it was concluded that the absence of external standardisation for the 25OHD assay might lead to greater variations and false results . To minimise the inter-laboratory and inter-method variations in our LC-MS/MS analysis, we have used reference materials for preparation of our calibrants and quality controls, along with the introduction of a new internal standard. The latter was required as, when using the internal standard hexadeuterated-25OHD3, certain fragment transitions of the parent ions could lead to greater isobaric interferences such as the transition 407.7 > 389.7 . Stanozolol-D3 as a new internal standard prevents isobaric interferences by using the transition 332.2 > 81.2.
To date, only one transition of the parent ion to product ion has been investigated, instead of our three-precursor ions to fragment ions transitions (according to the qualifier-qualifier principle) . To our knowledge, this is first time that three transitions of precursor ions to product ions have been used for quantification of 25OHD3, 25OHD2, 3-epi-25OHD3, 1αOHD3 and 7αC4 as shown in Table 2. The sum of the three transitions not only increases the sensitivity of the assay but also minimises isobaric interferences.
To summarise, this is the first LC-MS/MS method for the determination of 25OHD which includes separation and quantification of epimers and isobars. It is a fast liquid-liquid extraction method, which does not require complicated derivatisation procedures, hence reducing assay time and variability. This method can quantify the different 25OHD forms accurately, which is difficult to achieve using immunoassay methods. This methodology can be used for accurate blood testing to prevent falsely elevated Vitamin D levels being reported.
The LC-MS/MS method was free from all types of interferences arising due to epimeric, isobaric, instrumental and matrix components, which may interfere with analyte ionisation. Thus the LC-MS/MS method provides a robust, specific, reliable and reproducible technique as a solution to the problems identified in relation to current assays for Vitamin D. Removal of uncertainties in Vitamin D measurement, is required in order to progress the current understanding of the roles of Vitamin D in health and disease via more rigorous clinical trials .
The authors thank the volunteers who participated in this study. Authors also thank Dr Julian Swinden for his technical help.
- Dusso AS, Brown AJ, Slatopolsky E: Vitamin D. Am J Physiol Renal Physiol. 2005, 289: F8-F28. 10.1152/ajprenal.00336.2004.View ArticlePubMedGoogle Scholar
- Holick MF: Vitamin D deficiency. New Engl J Med. 2007, 357: 266-281. 10.1056/NEJMra070553.View ArticlePubMedGoogle Scholar
- Rajakumar K, Greenspan SL, Thomas SB, Holick FM: Solar ultraviolet radiation and Vitamin D: a historical perspective. Am J Public Health. 2007, 97: 1746-1754. 10.2105/AJPH.2006.091736.View ArticlePubMedPubMed CentralGoogle Scholar
- Holick MF: Sunlight and Vitamin D for bone health and prevention of autoimmune diseases, cancers, and cardiovascular disease. Am J Clin Nutr. 2004, 80 (6 Suppl): 1678S-1688S.PubMedGoogle Scholar
- Melamed ML, Muntner P, Michos ED, Uribarri J, Weber C, Sharma J, Raggi P: Serum 25-hydroxyvitamin D levels and the prevalence of peripheral arterial disease: results from NHANES 2001 to 2004. Arterioscler Thromb Vasc Biol. 2008, 28 (6): 1179-1185. 10.1161/ATVBAHA.108.165886.View ArticlePubMedPubMed CentralGoogle Scholar
- Holick MF: Vitamin D status: measurement, interpretation, and clinical application. Ann Epidemiol. 2009, 19: 73-78. 10.1016/j.annepidem.2007.12.001.View ArticlePubMedGoogle Scholar
- Zhang R, Naughton DP: Vitamin D in health and disease: Current perspectives. Nutr J. 2010, 9: 65-10.1186/1475-2891-9-65.View ArticlePubMedPubMed CentralGoogle Scholar
- Hollis BW: Circulating 25-hydroxyvitamin D levels indicative of Vitamin D sufficiency: implications for establishing a new effective dietary intake recommendation for vitamin D. J Nutr. 2005, 135: 317-322.PubMedGoogle Scholar
- Barragry JM, France MW, Corless D, Gupta SP, Switala S, Boucher BJ, Cohen RD: Intestinal cholecalciferol absorption in the elderly and in younger adults. Clin Sci Mol Med. 1978, 55: 213-220.PubMedGoogle Scholar
- Barragry JM, France MW, Boucher BJ, Cohen RD: Metabolism of intravenously administered cholecalciferol in man. Clin Endocrinol(Oxf). 1979, 11: 491-495. 10.1111/j.1365-2265.1979.tb03101.x.View ArticleGoogle Scholar
- Clemens TL, Zhou XY, Myles M, Endres D, Lindsay R: Serum Vitamin D2 and Vitamin D3 metabolite concentrations and absorption of Vitamin D2 in elderly subjects. J Clin Endocrinol Metab. 1986, 63: 656-660. 10.1210/jcem-63-3-656.View ArticlePubMedGoogle Scholar
- Zerwekh JE: The measurement of Vitamin D: analytical aspects. Ann Clin Biochem. 2004, 41: 272-281. 10.1258/0004563041201464.View ArticlePubMedGoogle Scholar
- Shimada K, Mitamura K, Higashi T: Gas chromatography and high-performance liquid chromatography of natural steroids. J Chromatogr A. 2001, 935: 141-172. 10.1016/S0021-9673(01)00943-8.View ArticlePubMedGoogle Scholar
- Carter GD, Carter CR, Gunter E, Jones J, Jones G, Makin HL, Sufi S: Measurement of Vitamin D metabolites: an international perspective on methodology and clinical interpretation. J Steroid Biochem Mol Biol. 2004, 89-90: 467-471.View ArticlePubMedGoogle Scholar
- Hollis BW: The determination of circulating 25-hydroxyvitamin D: no easy task. J Clin Endocrinol Metab. 2004, 89: 3149-3151. 10.1210/jc.2004-0682.View ArticlePubMedGoogle Scholar
- Horst RL, Reinhardt TA, Beitz DC, Littledike ET: A sensitive competitive protein-binding assay for Vitamin D in plasma. Steroids. 1981, 37: 581-591. 10.1016/S0039-128X(81)90403-7.View ArticlePubMedGoogle Scholar
- Jones G: Assay of Vitamins D2 and D3, and 25-hydroxyvitamins D2 and D3 in human plasma by high-performance liquid chromatography. Clin Chem. 1978, 24: 287-298.PubMedGoogle Scholar
- Terry AH, Sandrock T, Meikle AW: Measurement of 25-hydroxyvitamin D by the Nichols ADVANTAGE, DiaSorin LIAISON, DiaSorin RIA, and liquid chromatography-tandem mass spectrometry. Clin Chem. 2005, 51: 1565-1566. 10.1373/clinchem.2005.054239.View ArticlePubMedGoogle Scholar
- Carter GD, Carter R, Jones J, Berry J: How accurate are assays for 25-hydroxyvitamin D? Data from the international Vitamin D external quality assessment scheme. Clin Chem. 2004, 50: 2195-2197. 10.1373/clinchem.2004.040683.View ArticlePubMedGoogle Scholar
- Binkley N, Krueger D, Gemar D, Drezner MK: Correlation among 25-hydroxyvitamin D assays. J Clin Endocrinol Metab. 2008, 93: 1804-1808. 10.1210/jc.2007-2340.View ArticlePubMedGoogle Scholar
- Binkley N, Krueger D, Cowgill CS, Plum L, Lake E, Hansen KE, Deluca HF, Drezner MK: Assay variation confounds the diagnosis of hypovitaminosis D: a call for standardization. J Clin Endocrinol Metab. 2004, 89: 3152-3157. 10.1210/jc.2003-031979.View ArticlePubMedGoogle Scholar
- Hyppönen E, Turner S, Cumberland P, Power C, Gibb I: Serum 25-hydroxyvitamin D measurement in a large population survey with statistical harmonization of assay variation to an international standard. J Clin Endocrinol Metab. 2007, 92: 4615-4622. 10.1210/jc.2007-1279.View ArticlePubMedGoogle Scholar
- Roth HJ, Schmidt-Gayk H, Weber H, Niedrau C: Accuracy and clinical implications of seven 25-hydroxyvitamin D methods compared with liquid chromatography-tandem mass spectrometry as a reference. Ann Clin Biochem. 2008, 45: 153-159. 10.1258/acb.2007.007091.View ArticlePubMedGoogle Scholar
- Saenger AK, Laha TJ, Bremner DE, Sadrzadeh SM: Quantification of serum 25-hydroxyvitamin D(2) and D(3) using HPLC-tandem mass spectrometry and examination of reference intervals for diagnosis of Vitamin D deficiency. Am J Clin Pathol. 2006, 125: 914-920. 10.1309/J32UF7GTQPWN25AP.View ArticlePubMedGoogle Scholar
- Priego Capote F, Ruiz Jimenez J, Mata Granados JM, Luque de Castro MD: Identification and determination of fat-soluble Vitamins and metabolites in human serum by liquid chromatography/triple quadrupole mass spectrometry with multiple reaction monitoring. Rapid Commun Mass Spectrom. 2007, 21: 1745-1754. 10.1002/rcm.3014.View ArticlePubMedGoogle Scholar
- Binkley N, Ramamurthy R, Krueger D: Low Vitamin D status: definition, prevalence, consequences, and correction. Endocrinol Metab Clin North Am. 2010, 39: 287-301. 10.1016/j.ecl.2010.02.008.View ArticlePubMedPubMed CentralGoogle Scholar
- Higashi T, Yamauchi A, Shimada K: Application of 4-(4-nitrophenyl)-1,2,4-triazoline-3, 5-dione to analysis of 25-hydroxyvitamin D3 in human plasma by liquid chromatography/electron capture atmospheric pressure chemical ionization-mass spectrometry. Anal Sci. 2003, 19: 941-943. 10.2116/analsci.19.941.View ArticlePubMedGoogle Scholar
- Higashi T, Homma S, Iwata H, Shimada K: Characterization of urinary metabolites of Vitamin D(3) in man under physiological conditions using liquid chromatography-tandem mass spectrometry. J Pharm Biomed Anal. 2002, 31: 947-955.View ArticleGoogle Scholar
- Vogeser M: Quantification of circulating 25-hydroxyvitamin D by liquid chromatography-tandem mass spectrometry. J Steroid Biochem Mol Biol. 2010, 121: 565-573. 10.1016/j.jsbmb.2010.02.025.View ArticlePubMedGoogle Scholar
- Maunsell Z, Wright DJ, Rainbow SJ: Routine isotope-dilution liquid chromatography tandem mass spectrometry assay for simultaneous measurement of the 25-Hydroxy metabolites of Vitamins D2 and D3. Clin Chem. 2005, 51: 1683-1690. 10.1373/clinchem.2005.052936.View ArticlePubMedGoogle Scholar
- Ginty F, Rennie KL, Mills L, Stear S, Jones S, Prentice A: Positive, site-specific associations between bone mineral status, fitness, and time spent at high-impact activities in 16- to 18-year-old boys. Bone. 2005, 36: 101-110. 10.1016/j.bone.2004.10.001.View ArticlePubMedGoogle Scholar
- Eyles D, Anderson C, Ko P, Jones A, Tomas A, Burne T, Mortensen PB, Norggard-Pederser B, Hougaard DM, McGrath J: A sensitive LC/MS/MS assay of 25OH Vitamin D3 and 25OH Vitamin D2 in dried blood spots. Clin Chim Acta. 2009, 403: 145-151. 10.1016/j.cca.2009.02.005.View ArticlePubMedGoogle Scholar
- Higashi T, Awada D, Shimada K: Simultaneous determination of 25-hydroxyvitamin D2 and 25-hydroxyvitamin D3 in human plasma by liquid chromatography-tandem mass spectrometry employing derivatization with a Cookson-type reagent. Biol Pharm Bull. 2001, 24: 738-743. 10.1248/bpb.24.738.View ArticlePubMedGoogle Scholar
- Higashi T, Awada D, Shimada K: Determination of 24,25-dihydroxyvitamin D(3) in human plasma using liquid chromatography-mass spectrometry after derivatization with a Cookson-type reagent. Biomed Chromatogr. 2001, 15: 133-140. 10.1002/bmc.43.View ArticlePubMedGoogle Scholar
- Gao S, Zhang ZP, Karnes HT: Sensitivity enhancement in liquid chromatography/atmospheric pressure ionization mass spectrometry using derivatization and mobile phase additives. J Chromatogr B. 2005, 825: 98-110. 10.1016/j.jchromb.2005.04.021.View ArticleGoogle Scholar
- Byrdwell WC: Dual parallel mass spectrometry for lipid and Vitamin D analysis. J Chromatogr A. 2010, 25: 3992-4003.View ArticleGoogle Scholar
- Chen H, McCoy LF, Schleicher RL, Pfeiffer CM: Measurement of 25-hydroxyvitamin D3 (25OHD3) and 25-hydroxyvitamin D2 (25OHD2) in human serum using liquid chromatography-tandem mass spectrometry and its comparison to a radioimmunoassay method. Clin Chim Acta. 2008, 391: 6-12. 10.1016/j.cca.2008.01.017.View ArticlePubMedGoogle Scholar
- Bischof MG, Siu-Caldera ML, Weiskopf A, Vouros P, Cross HS, Peterlik M, Reddy GS: Differentiation-related pathways of 1,25-dihydroxycholecalciferol metabolism in human colon adenocarcinoma-derived Caco-2 cells: production of 1,25-dihydroxy-3epi-cholecalciferol. Exp Cell Res. 1998, 241: 194-201. 10.1006/excr.1998.4044.View ArticlePubMedGoogle Scholar
- Siu-Caldera ML, Sekimoto H, Weiskopf A, Vouros P, Muralidharan KR, Okamura WH, Bishop J, Norman AW, Uskokovic MR, Schuster I, Reddy GS: Production of 1,25-dihydroxy-3-epi-vitamin D3 in two rat osteosarcoma cell lines (UMR 106 and ROS 17/2.8): existence of the C-3 epimerization pathway in ROS 17/2.8 cells in which the C-24 oxidation pathway is not expressed. Bone. 1999, 24: 457-463. 10.1016/S8756-3282(99)00019-8.View ArticlePubMedGoogle Scholar
- Sekimoto H, Siu-Caldera ML, Weiskopf A, Vouros P, Muralidharan KR, Okamura WH, Uskokovic MR, Reddy GS: 1,25-Dihydroxy-3-epi-vitamin D3: in vivo metabolite of 1,25-dihydroxyvitamin D3 in rats. FEBS Let. 1999, 448: 278-282. 10.1016/S0014-5793(99)00377-4.View ArticleGoogle Scholar
- Singh RJ, Taylor RL, Reddy GS, Grebe SKG: C-3 epimers can account for a significant proportion of total circulating 25-hydroxyvitamin D in infants, complicating accurate measurement and interpretation of Vitamin D status. J Clin Endocrinol Metab. 2006, 91: 3055-3061. 10.1210/jc.2006-0710.View ArticlePubMedGoogle Scholar
- Kamao M, Tatematsu S, Sawada N, Sakaki T, Hatakeyama S, Kubodera N, Okano T: Cell specificity and properties of the C-3 epimerization of Vitamin D3 metabolites. J Steroid Biochem Mol Biol. 2004, 89-90: 39-42.View ArticlePubMedGoogle Scholar
- Kamao M, Tatematsu S, Hatakeyama S, Sakaki T, Sawada N, Inouye K, Ozono K, Kubodera N, Reddy GS, Okano T: C-3 epimerization of Vitamin D3 metabolites and further metabolism of C-3 epimers: 25-hydroxyvitamin D3 is metabolized to 3-epi-25-hydroxyvitamin D3 and subsequently metabolized through C-1 or C-24 hydroxylation. J Biol Chem. 2004, 279: 15897-15907. 10.1074/jbc.M311473200.View ArticlePubMedGoogle Scholar
- Carter GD: 25-hydroxyvitamin D assays: the quest for accuracy. Clin Chem. 2009, 55: 1300-1302. 10.1373/clinchem.2009.125906.View ArticlePubMedGoogle Scholar
- Vogeser M, Kyriatsoulis A, Huber E, Kobold U: Candidate reference method for the quantification of circulating 25-hydroxyvitamin D3 by liquid chromatography-tandem mass spectrometry. Clin Chem. 2004, 50: 1145-1147.View ArticleGoogle Scholar
- Lewis JG, Elder PA: Serum 25-OH Vitamin D2 and D3 are stable under exaggerated conditions. Clin Chem. 2008, 54: 1931-1932. 10.1373/clinchem.2008.111526.View ArticlePubMedGoogle Scholar
- Holick MF: Environmental factors that influence the cutaneous production of Vitamin D. Am J Clin Nutr. 1995, 61 (3 Suppl): 638S-645S.PubMedGoogle Scholar
- Joint position statement issued to provide vitamin D clarity. [http://info.cancerresearchuk.org/news/archive/cancernews/2010-12-16-Joint-position-statement-issued-to-provide-vitamin-D-clarity-?rss=true]
- Holick MF: Vitamin D status: Measurement, interpretation, and clinical application. Ann Epidemiol. 2009, 19 (2): 73-78. 10.1016/j.annepidem.2007.12.001.View ArticlePubMedGoogle Scholar
- Dietary Supplement Fact Sheet: Vitamin D. [http://ods.od.nih.gov/factsheets/vitamind]
- Pearce SHS, Cheetham TD: Diagnosis and management of vitamin D deficiency. BMJ. 2010, 340: b5664-10.1136/bmj.b5664.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.